TM 8-612 Distilling Apparatus 1943
, Technical Manual
, Field Manual
, World War II
, World War, 1939-1945 -- Medical Care -- United States
, World War, 1939-1945
, Distillation apparatus -- United States -- Maintenance and repair -- Handbooks, manuals, etc
, Surgical instruments and apparatus -- United States -- Handbooks, manuals, etc
, United States. -- Army -- Medical care -- Handbooks, manuals, etc
, United States. Army
, Distillation apparatus
, Surgical instruments and apparatus
United States War Department Technical Manual TM 8-612 Distilling Apparatus. 1943-07-13
Collection wwIIarchive; additional_collections
Full catalog record MARCXML
PART ONE. INTRODUCTION.
Section I. General.
Section II. Description and tabulated data.
PART TWO. OPERATING INSTRUCTIONS.
Section III. General.
Scope 6 8
Section IV. Service upon receipt of equipment.
Unpacking and assembling, Market Forge
Models 7 8
Unpacking and assembling, Consolidated
Machine Works Models 8 8
Section V. Controls and instruments.
Controls 9 9
Instruments 10 9
Section VII Operation
General 11 10
Operation, steam-heated model 12 10
Operation, gasoline-heated model 13 10
PART THREE. MAINTENANCE INSTRUCTIONS.
Section VII. General.
Scope 14 20
Section VIII. Preventive maintenance services.
Operator maintenance (First Echelon) 15 20
Organizational maintenance (Second Echelon) 16 20
Section IX. Trouble shooting.
If the apparatus fails to produce distilled water in rated quantities 17 21
Section X. Maintenance operations.
Cleaning boiler 18 21
Boiler fails to heat properly, steam model 19 21
Boiler fails to heat properly, gasoline model 20 22
Steam leak between evaporator and boiler cover 21 22
Cleaning and replacing glass gauge, 5-gallon units 22 22
APPENDIX. SHIPMENT AND STORAGE
(DISASSEMBLING AND PACKING) 24
Digitized by Google
This book has an
editable web page